Download HP A RF IMPEDANCE-ANALYZER OPERATION SM service manual & repair info for electronics experts. The HP A RF Impedance Analyzer measures 14 parameters Parameter measured: IZI, Swept measurement: manual or automatic sweep from start to. Agilent A Test Fixture. Operation and Service Manual. Third Edition. Agilent Part No. February Printed in Japan.
|Published (Last):||25 July 2008|
|PDF File Size:||6.48 Mb|
|ePub File Size:||20.82 Mb|
|Price:||Free* [*Free Regsitration Required]|
Message 6 of 6. It’s 84MB, so I hope you have a fast link. Accepted by topic author Matt C.
Sorry, that last line of code there is supposed to read: The forums emoticons took over for a bit there. Message 3 of 6. Guess you never know until you try! Thanks for your help guys.
4191A RF IMPEDANCE ANL
If not, what would creating a driver for it entail? I am able to communicate with the device through the GPIB interface, and am also able to retrieve information. Would you mind posting the code you used on the A?
Message 1 of 6. Most Active Software Boards: Good luck with it Matt – you’re right, the commands for the A are completely different from the – probably because of its age.
Solved: GPIB Coding for the HPA Impedence Analyzer – Discussion Forums – National Instruments
I went through the manual before, but when I saw their GPIB section, the strings were so convoluted and so unlike anything I’ve mwnual seen before that I assumed it was referring to something different. You really need the manual. I suspect that this difference might be causing my problems.
I have worked with the A and, as I recall, the commands are nothing like any other instrument I have seen. In the code above, MHz is an arbitrary frequency. IIRC, the sweep parameters were something like this: I think the would probably take the old commands, but the ALSO has new commands.
In particular, after setting up and initiating the instrument, Manal am initiating the measurement and reading it as follows: Message 4 of 6. Should I be looking at doing it with LabView or something else?
When I gave it a try, though, it worked fine. Message 5 of 6. Has anyone done anything on interfacing with the A since Matt C’s last post?
Keysight / Agilent A – In Stock, We Buy Sell Repair, Price Quote
Message 2 of 6. Manuxl helps you quickly narrow down your search results by suggesting possible matches as you type. In particular, after setting up and initiating the instrument, I am initiating the measurement and reading it as follows:.
I have not worked with that instrument, but the Agilent web site has the operating and service manual here.
However, rather than being returned the data at MHz, I am returned the data at whatever frequency the analyzer is currently set at.