to reading the User’s Manual. Please read the Accutrend. Plus User’s Manual carefully before using the meter for the first time. The Accutrend Plus meter is used. The Accutrend Plus system, one meter delivering both total cholesterol and glucose test results, can with Accutrend Plus may help employers improve employee productivity, improve .. 15 Cholestech LDX System User Manual. Inverness. View and Download Cobas Accutrend Plus user manual online. Accutrend Plus Blood Glucose Meter pdf manual download.
|Published (Last):||22 July 2016|
|PDF File Size:||15.95 Mb|
|ePub File Size:||1.1 Mb|
|Price:||Free* [*Free Regsitration Required]|
Operating the instrument Ensure the instrument is switched off and turn it over. Memory Memory The Accutrend Plus instrument has two memory areas, each of which can be used to store up plud measured values together with date, time and event flags. You need a free hanging blood drop for measurement.
Cobas Accutrend Plus User Manual
Check that the whole display is functioning correctly to prevent misinterpretations due to defective display elements. Each time you press the Set button, the last stored value of each memory area, if any, is displayed.
When you press the button, the place of the measured value within the memory is displayed here: Performing A Control Test Quality control recommendation Performing a control test Place the instrument on a level surface or hold it in your hand. Accutend messages Display Error meaning and description Error: Do not lance yourself before being instructed to do so in the course of this description. Page 90 Quality control recommendation The measurement time varies depending on the test parameter.
Display And Symbols The Accutrend Plus instrument Display and symbols Mabual time the instrument is powered on, you should check the display. Quality control recommendation Open the measurement chamber flap. Page 87 Quality control recommendation Open the measurement chamber flap. The Accutrend Plus instrument is used for quantitative measurement of two blood parameters: The display then automatically switches to the mode for setting the time format.
Hold the test strip with thumb and index finger so that the printed measurement parameter is facing upwards. The packaging material, the identification plate of the instrument and the manual may contain the following symbols or abbreviations which are listed below with their meaning: Don’t show me this message again.
COBAS ACCUTREND PLUS USER MANUAL Pdf Download.
Performing a measurement Hold the test strip with thumb and index finger so that the sample application area is facing upwards. Insert the code strip smoothly into the test strip guide up to the stop in the direction of the printed arrows.
Switching On The Instrument Performing a accuttend Switching on the instrument Place the instrument on a level surface or hold it in your hand. The next time the Set button is pressed, the minutes can be set again with the M button. Instrument settings Press the Set button again to save the current setting and go to the next setting. Page 99 Cleaning If needed, you can rinse the strip guide separately from the instrument under manyal running water.
Page 40 Instrument settings The default day flashes. Target values which should be obtained can either be found on the labels or on the package inserts for the test strips or control solutions.
Performing a measurement Alternative blood application outside the device: Instrument settings Setting the date With the plud three settings, you first enter the year, then the month and finally the day. Performing a measurement Blood collection Lance the outer side of the fingertip using the lancing device to obtain a large hanging drop of blood.
Lift llus lid upwards to remove it from the instrument. Page 35 Instrument settings If the displayed setting is correct e. Flap open displayed after instrument has been powered on The measurement chamber flap is not closed properly. Display Of Results Quality control recommendation Display of results When measurement is finished, the result is displayed.
Page 36 Instrument settings Press the Set button again to save the current setting and go to the accutrens setting. Alphabetical Index Alphabetical index Alphabetical index Capillary blood Press the M button to change the currently flashing figure and to set the current year.
You can only move forward through the settings. Introduction Introduction The Accutrend Plus instrument The Accutrend Plus instrument is used for quantitative measurement of 2 blood parameters: Performing a measurement Now take the test strip from the test strip container. The application area must be completely covered with blood example Aotherwise incorrect values may be obtained. Contents Product specifications Operating conditions and technical data